Response to fixed CIS signal for 48 channels (high gain). Three channels from one digitizer Tile DMU chip have a
large pedestal and three channels are unused. The labeling of two unused channels is shifted by one because of
a minor firmware problem.
System noise for 48 channels. The "high frequency noise" corresponds to the RMS of the 9 pedestal samples within
an event. The "low frequency noise" corresponds to the pedestal variation from event to event.
Simulated radiation levels over 10 years at z=150 cm, as of 11/18/00
Total ionizing dose 0.23 Gy (0.023 Krad)
Non-ionizing radiation 1.5 x 1010 1-MeV-equiv-neutrons/cm2